The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||6 March 2014|
|PDF File Size:||16.14 Mb|
|ePub File Size:||14.30 Mb|
|Price:||Free* [*Free Regsitration Required]|
Turn-On Checkout – 4Procedure6.
Providing the finest test equipment solutions since 1992
Selftest ErrorsPressing the Shift, Error keys will show the error number. Page 35 5Front panel OperationIntroductionHere is what you will find in this chapter: Turn off the dc source and connect it as shown in figure B-1Bwith the ohm load resistor.
Alexa Actionable Analytics for the Web. CopyleftAll Wrongs Reserved. Page 75 Manual UpdatesThe following updates have been made to this manual since the November printingindicated on the Printing History page. Table B-1 lists the equipment required for calibration.
HP Agilent Keysight, B
Were they so fancy they were using 4-wire? Page 33 Turn-On Checkout – 4Procedure6. Your manual failed to upload Wait for the DMM readingto stabilize. Refer to Table to calculate the voltage dropfor some commonly used AWG copper wire. Page 26 3 – Installationat the load.
Set the instrument to the operating state that you want to save. Press the OCP key toturn over-current protection off. Page 22 3 – InstallationLocationThe outline diagram in figure gives the dimensions of your dc source. If looking through the HP Hewlett-Packard B user manual directly on this website is not convenient for you, there are two possible solutions:.
Amazon Restaurants Food delivery from local restaurants. How you did the voltage calibration? I just got it from a good friend of mine, who owned two of these fine instruments.
Get fast answers from reviewers.
HP (Hewlett-Packard) 6632B user manual
Enter the new password from the keypad. UnrThe output is unregulated. On the Function keypad, press Protect. Unless otherwise noted, specifications apply when measured atthe rear terminals after a minute warm-up period.
After some head scratching and a LANG change this was resolved. HP Hewlett-Packard B user manual. ERROR where “n” is 663b number listed in the following table. The option to print the manual has also been provided, and you can use it by clicking the link above – Print the manual. Parameter lists are circular; you can return to the startingposition by continuously pressing either key.
HP B : – Hewlett-Packard Power Supply User Manual
United States of America: Readings within high currentlimits see tables B-3,4, 5 or 6. The verification procedures do not check all the operating parameters, but verify that the dc sourceis performing properly. Failure to comply with these precautions or with specific warnings elsewhere in this manual violates safetystandards of design, manufacture, and intended use of the instrument.
It includes a dc source, whichproduces dc output with programmable voltage and current amplitude, and a highly accuratevoltage and current meter, with the capability to measure very low-level currents. Front Panel DescriptionFigure Pressthe Local key on the front panel to put the unit in local mode.
In this mode the portfunctions as a remote inhibit input with a discrete fault indicator ph signal.