Keyboard Layout v Fluke B/1 96B-C/B-C Service Manual List of Figures For the mA. current D pin 3 is connected to pin 1, and pin 13 is. Manual de Servicio Fluke b – Ebook download as PDF File .pdf), Text File . txt) or trigger level mV) or (R the switches in D are all open. F HP Deskjet F HP Deskjet D HP Deskjet D HP Deskjet D . would work while others would not, and maybe it’s some strange fluke.
|Published (Last):||25 March 2016|
|PDF File Size:||5.76 Mb|
|ePub File Size:||15.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
Adjust the A output voltage until the displayed trace has the trigger amplitude indicated under the last column of Table Remove the two screws item 14 that secure the Main PCA unit to the top case. Observe the main reading and check to see if it is within 0.
Use the following procedure to replace the strap, or to change the strap position. Pull sideward d156 the front edge of the tilt stand until the hinge releases. It provides a quick way to get familiar with the test tool basic build-up. Turn the test tool on, and monitor the voltage on T From the measured voltage and supplied current the resistance value is calculated.
HP C9352AE (22) Multi Colour Cartridge for Inkjet Printer
The output data and clock are supplied d160 pulse shapers D If one or more currents OK then check D and control signals. The PLL is synchronized with the external crystal B The information in this manual was current when published.
Check N 1 The test tool is operative now.
Optional Accessories 9- 1. Make sure that the tabs on the Shielding D560 go into the slots in the top of the Top Case. Set the tilt stand to a 45 -degree position respective to the test tool bottom.
Remove the BC adapter power, and verify that the backlight is dimmed.
Full text of “FLUKE B, B, C, B,C Service”
If linearization fails during calibration, then select calibration mode O-O. A contaminated PCA may not cause immediate instrument failure in controlled environments. X Series Application Note Probe calibration By switching a current on and off, a Hz square wave for probe calibration is generated.
Flukee not OK see section 7. Then the self oscillating Fly Back Converter becomes active. If the instrument is on, it must alternately show the voltage levels on D pins 5, 12, 14and 15; if at a fixed level, then replace D If the voltage drops below approximately 1.
Full text of “FLUKE “
Probe dc calibration can only be done if probe Check the HF path: The following performance tests are provided to ensure that the test tool is in a proper operating condition. Required voltages on flukke test points: Share Facebook Twitter Pinterest. Continue through all calibration points. The technical specifications and operating guides included in this manual More information. Check D1 57 1 pin 2 for positive pulses.
Apply the inverted TV Signal Generator signal to the test tool.
Share your thoughts with other customers. To locate an authorized service center refer to the second page of this manual back of the title page.
Reattach the LCD flex cable, secure that cable in the connector with the connector latch. Willis Patterson 1 years ago Views: Proceed as follows to test the Peak measurement accuracy: