When the e.m.f. being measured is equal to the potentiometer voltage, the amplifier output Since polar potentiometer was developed by Dr. C.V. Drysdale , it is. Dr. C.V. Drysdales Alternating and Continuous Current Potentiometer The instrument comprises the standard Tinsley DC potentiometer of the time, with a phase. In a.c. potentiometers, another requirement is the control of stray e.m.f.s in fields Being a polar type, Drysdale – Tinsley a.c. potentiometer measures unknown.
|Published (Last):||3 July 2011|
|PDF File Size:||5.70 Mb|
|ePub File Size:||14.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
It has electrodynamometer type ammeter along with DC potentiometer and phase-shifting transformer which is operated by single phase supply.
There are two types of AC potentiometers: These are adjusted until the A. Now to measure unknown e. In a phase-shifting transformer, there is a combination of two ring-shaped laminated steel stators connected perpendicularly to each other as shown in the figure. There are two step-down transformers T 1 and T 2 which isolate potentiometer from the line and give an earthed screens protection between the winding.
The known source may be DC or AC. There is a provision on the scale that it could read phase angle up to This is done by switching it over on to a continuous current circuit, which is adjusted until a balance is obtained at the proper setting, when a Weston cell is connected up in the usual manner. The main circuit of the potentiometer must be supplied with alternating current of exactly the same frequency as that in the circuit to be measured, which means in practice that they must be supplied from the same source.
Now the in-phase component V A of the unknown e. Drysdale’s Alternating and Continuous Current Potentiometer The instrument comprises the standard Tinsley DC potentiometer of the time, with a phase shifter, Weston dynamometer voltmeter, selection and change-over switches and terminal panels for connection of galvanometers, DC and AC measuring and supply voltages.
Since opening at the end ofover 5, children have visited the Centre.
The whole arrangement of the winding is done in such a way that the magnitude of the induced emf in the rotor may change but it does not affect the phase angle and it can be read on the scale fixed on the top of the instrument. Further, the phase relation between the circuit to be measured and that supplying the potentiometer may have any value from to But there is one major difference between their measurements, DC potentiometer only measures the magnitude of the unknown voltage.
Measures voltage as magnitude and phase from reference voltage. The potentiometer, which is the universal instrument by which currents are determined in terms of a difference of potential between the end of a known resistance, has been developed by C.
The main potentiometer current must be kept at some known constant value. Web Design by Pure Energy. Drysdale for the measurement of alternating potentials.
These children deserve a space which is engaging and instructive, where they can feel a sense of adventure, exploration and surprise! The working phenomenon of DC potentiometer and AC potentiometer is same. As it is necessary to have the splitphase circuit somewhere near the resonating point, it must be adjusted for changes of frequency. Arrangements are provided for reversing the ammeter in order to eliminate the effect of stray fields.
Join our Mailing List by adding polae email address below and be kept up to date with lots of information including:.
This ammeter must read correctly with alternating and continuous currents. S 1 and S 2 are signs changing switches which are used to change the polarity of the test voltage if it is required for balancing the Potentiometer. To obtain the necessary field distribution a split-phase scheme is used, part rrysdale the excitation being provided by a circuit containing a condenser and a resistance in series.
Please be assured that any amount you feel able to donate will make an immediate and significant impact. The first one potnetiometer named as the in-phase potentiometer which is used to measure the in-phase factor of an unknown e. We are extremely grateful for the very kind donation from Rosemary Helme of this Drysdale Alternating and Continuous Current Potentiometer.
The instrument comprises the standard Tinsley DC potentiometer of the time, with a phase shifter, Weston dynamometer voltmeter, selection and change-over switches and terminal panels for connection of galvanometers, DC and AC measuring and supply voltages. Polar type Potentiometer In such type of instruments, two separate scales are used to measure magnitude and phase angle on some reference of the unknown e.
It also supplies 6 volts to potentiometers. The phase displacement of the rotor emf is equal to rotor movement angle from its original position and it is related to the stator supply voltage.
Drysdale ac polar potentiometer
Connects Facebook Youtube Videos. Further references can be found at: A rotation of the rotor of the phase-shifting transformer through o has the effect of producing a phase shift of the same value, and a pointer on the rotor axis indicates the phase angle on a suitably divided scale. The rotor has a winding on it which supplies the potentiometer current.
Coordinate type Potentiometer In coordinate AC potentiometertwo ootentiometer potentiometers are caged in one circuit as shown in the figure.